Structure of the PC measurement electronics with Spider8. B-1 .. When exchanging any modules, only original HBM parts must be used. Qualified personnel. Spider8 – Easy and reliable PC-based The Spider8 is a multi-channel PC measurement electronics for parallel, dynamic . Our factory in Darmstadt or HBM. Catman talks to Spider8 remotely. Keyword: Wireless. Remote communication with field equipment up to 5km away is now possible with HBM’s improved.
|Published (Last):||10 August 2017|
|PDF File Size:||7.18 Mb|
|ePub File Size:||5.49 Mb|
|Price:||Free* [*Free Regsitration Required]|
Perhaps the old firmware caused some problems Are you getting spidsr8 error messages from the VI’s or does the device just not seem to respond at all?
A big problem were the error messages which sometimes didn’t make sense to me. I would like to receive help with respect this issue.
Lot 753: HBM Spider 8 Data Acquisition Panels
First at all say that my spider8 is version P13 quite old. Message 1 spiver8 For Labview there are some examples which show how to access data via catman activeX. My OS is windows 98, labview version 6i. Any advice is welcome as well as was a place where I still can download the last firmware version. I hope this help you.
I guess, the best way is contacting a sales representative of HBM http: Message 10 of I have read that minimum version shoud be P20 to work correcrtly with Labview. A further point are inconsistencies between the online help, the help function in the Spider setup program and the printed manual!
Now I have got problems with data acquisition, it doesn’t work at all.
LabView and Spider8 from HBM – Discussion Forums – National Instruments
The big advantage is, that you have all options to control your DAQ process. Best Regards Andreas R. Message 2 of I am reading your inputs with respect the use of the Spider8 with Labview.
Message 6 of First of all, my operating system is NT4. Message 7 of So I will give some remarks on what could have been the reason: Message 9 of Message 3 of HBM home page doesn? The cable has got a length of about 1 meter.
This Tutorial has information regarding how to properly call functions from DLLs using the call library node. So in this example, the timeout has to be adjusted in respect to the measuring rate. There is one little disadvantage: I have run the firmware sp8firm32 and de archive sp32, both can be downloaded in the hbm website, and my example started to run. I didn’t see any drivers for the spider8 on the NI site. In most cases the baud rate will be the limiting factor, but don’t forget the measuring rate!
Also, which drivers are you using exactly? Spider32 setup still can read it but seems a lot slower now. I solved the problems a few days ago.
I have updated everything. This is not very useful! Message 4 of I cannot tell exactly why the data acquisition process didn’t work, but now everything is fine.
Message 8 of May be, he has a demo version for you almost catman professional including the catman activeX interface.
I downloaded the LabView driver for this measurement device from the NI homepage. W henever this is not true, the Spider should be turned off and on.